The detection value of C.trachomatics should be higher than 2x103 IFU/ml
The detection value of C.trachomatics should be higher than 5x104 IFU/ml
Compared with 8 different kinds infection factors and the test coincidence rate is 8/8
Intend to useNew! Chlamydia +Gonorrdea CT&NG Combo rapid test kit
The Chlamydia trachomatics antigen test kit is a rapid visual immunoassay for the qualitative detection of Chlamydia trachomatics in female endocervical swab, male urethral swab and urine specimens. This kit is intended for use as an aid of diagnosis of Chlamydia trachomatics infection.
The Neisseria Gonorrheaantigen test kit is a rapid visual immunoassay for the qualitative detection of Neisseria Gonorrhea in female endocervical swab and male urethral swab specimens. This kit is intended for an aid in the diagnosis of Neisseria Gonorrheainfection.
Operating StepsNew! Chlamydia +Gonorrdea CT&NG Combo rapid test kit
Prepare endocervical or Urethral swab specimens:
1)Add 5-6drops (approximately 200µl) of Lysis solution A into the extraction tube vertically. Immerse the specimen swab into the extraction tube. Squeeze the tube and twirl the swab for 15 times. Then keep the swab in the extraction tube for 2 mins
2)Add 5-6drops (approximately 200µl) of Lysis solution B into the same extraction tube.Deposit may be produced in the mixture. Squeeze the tube and twirl the swab for 15 times. Then keep the swab in the extraction tube for one minute. The solution will be brown if the swab with blood. Wring all extractive from swab and discard the swab. Finally, cover the extraction tube with a filtration tip.
Prepare urine specimen
Add 5-6 drops (approximately 200µl) of Lysis solution B into the tube. Shake to mix well (use vortex mixer if necessary). Transfer the mixture from centrifugal tube to extraction tube, add 5-6 drops (approximately 200µl) of Lysis solution A, and squeeze the wall of tube 15 times and stay for one minute.
Dispense 2 drops of extracted samples from extraction tube to the both two sample well(S) on the test cassette, read the result within 10-15 minutes. Do not interpret the result after 15 minutes
1. Positive Coincidence Rate:100%
2. Negative Coincidence Rate:100%
3. Analytical Sensitivity:The detection value of Gonorrhea antigens should be higher than 5x104 in each specimen. The detection value of C.trachomatics should be higher than 2x103 IFU/ml 4. Analytical Specificity: Compared with difference kinds infection factors and the test result are negative.
5. Intra-batch Discrepancy: The test results are the same in one batch
6. Inter-batch Discrepancy: The test results are the same in different batch
Storage and stability
The kit should be stored at 2-30°C in a cool and dry place, protected from light
The validity duration is 12 months
Do not freeze.
Do not use after the expiration date indicated on the package.
Packaging & Shipping
Sexually Transmitted disease Tests
Detect N.G infection by Enzyme Reaction
urine/Cervical secretion/urethral discharge
N.G Ag test cassette
Endocervical or urethral secretion
CT Ag test cassette
Endocervical or urethral secretion
Detect BV based on four indicator:H2O2 ,SA,LE,PH
Detect BV based on polyamine test
Detect BV based on sialidase activity test
Detect BV based on Six indicator:H2O2 ,SA,LE,PH,CV,TV)
Human Immunodeficiency Virus (HIV 1/2) test
Human Immunodeficiency Virus (HIV 1+O/2) test cassette
HSV-2 Rapid Test
Rubella IgG/IgM Rapid Test
Other Series (If interested please click the title)